Interested in linking to "When Is It Time to Retire Your PAT System?"?
You may use the Headline, Deck, Byline and URL of this article on your Web site. To link to this article, select and copy the HTML code below and paste it on your own Web site.
By Gary D. Nichols, PE
This is the tenth article in this series of Controlglobal articles on process analyzer systems, and the third and last on process analyzer system reliability, maintenance and service. (See PAT System Reliability and Process Analyzer Reliability, Maintenance & Service.) Inasmuch as these articles cover the lifetime costs of process analyzer systems, we must address the wear-out and obsolescence of analyzer systems.
While it would seem at first glance that dealing with wear-out and obsolescence would be a simple matter of replacing a necessary analyzer system and permanently removing analyzers that are not in use, for whatever reason, we shall see that the matter is not quite so simple in every case. Financial and regulatory considerations supplement technological wear-out and obsolescence factors in the decision to remove and/or replace analyzer systems—or not.
Financial justification is usually the reason for removing an analyzer system from service. One financial justification might be that service and maintenance costs have risen above the cost of capital or annual depreciation for the replacement. However, recall from previous articles and personal experience that replacement analyzer systems may be more costly than expected due to safety, ergonomic, environmental and technological improvements to the analyzer system since its original installation.
You might also want to remove an analyzer system from service because of poor performance or because you need the physical space for other uses. The system could also be an experimental installation that did not meet expectations or one that is inherently is out of safety or environmental compliance. To understand these justifications, it is helpful to remember that an analyzer system installation typically has a lifetime of 10 to 20 years, during which any of these justifications might arise.
Technological improvements which result in improved safety and environmental considerations, better accuracy or precision, up-to-date communications modes, new analytical methods, more ruggedness, improved stability to power-line upsets, lower service and maintenance costs, etc., may justify replacing an analyzer system. Analyzer systems seldom are replaced precipitously for these reasons, but technological improvements may accelerate a replacement schedule.
Lastly, regulatory requirements may necessitate replacing an analyzer system. Personnel protection, loss prevention or environmental regulations can all motivate replacement. Personnel protection issues could include the ways sample or calibration fluids are vented and disposed of, exposed hot or cryogenic surfaces or electrical wires, poor ergonomics, and dangerous (glass) sample-handling components. Experimental analyzer installations are usually in a special expensed class of assets.
Usually, replacement or removal is sold based a combination of financial, technological and regulatory. Although some of the reasons may be difficult to put in financial terms, financial justifications generally lead the way. Again, this is a good time for the analyzer engineer to consult the capital equipment accountant and the operations manager responsible for the analyzer.
Finally, let us become familiar with some of the terms that are helpful when discussing analyzer system wear-out and obsolescence. These terms probably are defined in engineering project cost accounting. They are usually expensed items, but the analyzer project engineer should check with the capital equipment accountant or cost engineer regarding their proper use as capital or expense for the specific project.
ControlGlobal.com is exclusively dedicated to the global process automation market. We report on developing industry trends, illustrate successful industry applications, and update the basic skills and knowledge base that provide the profession's foundation.