Greg McMillan and Stan Weiner bring their wits and more than 66 years of process control experience to bear on your questions, comments, and problems. Write to them at email@example.com.
By Greg McMillan and Stan Weiner
Greg: What we ultimately want to know and control is the composition of a stream. The lack of analyzers to tell us this is the primary limitation to more advanced process control and optimization (See my blog post, "Top Ten Limitations – Analyzers"). I have connections with a strong ISA analysis division through my pH expertise. I recently interviewed Jim Tatera, the analyzer track chair. Jim retired from a long career as an analyzer specialist at a very large chemical company and is now the Senior Process Consultant at Tatera & Associates, Inc. Jim started out in analytical research labs and moved into online applications after developing an appreciation of the opportunities in production from discussions with operations and process support personnel. We are fortunate to have Jim and his colleagues help us appreciate the rich history and future of process analyzers over the next several months.
Stan: We had quite a strong group of analyzer specialists in the 1960s and 1970s in our large chemical intermediates plants with continuous processes for monomers and polymers. The batch processes had fewer analyzer applications. Newer agricultural batch processes with higher value-added products had slightly more analyzers than specialty chemical batch processes. Mature batch processes relied almost entirely on lab analysis.
Greg: I just talked to an analyzer specialist at a nylon plant I supported in Florida and found that there was only one analyzer specialist left. I asked why. The remaining specialist said the number of applications has stayed about the same, but they are adequately supported by four technicians. The peak in analyzer specialists in the 1970s was the result of the need to develop, design and build analyzer systems including circuit boards. Now these analyzers are available as complete packages from analyzer suppliers with integrated smart microprocessors. Were we a special case?
Jim: Most of the mature technology today for analyzers was developed at chemical plants and refineries by the 1970s. End users designed, constructed and assembled the components, enclosures, sample systems, circuits and interfaces. The analyzer technology was subsequently licensed to today's analyzer suppliers when the process industries in the 1980s decided oil, gas and chemical production, not analyzer manufacturing was their core competency. In some cases the licensing of technology included a 50% ownership of shares in the original analyzer companies. Many of these companies were then bought by large automation system suppliers. The numbers of analyzer specialists have dramatically declined since complete packages can be purchased that are capable of being supported by technicians. One concern is that the few specialists left onsite or as retired consultants may not be around in 10 years, stifling new analyzer applications.
Stan: How did you find new analyzer opportunities?
Jim: We had temporary relocate-able process analyzers (TRPA) that we got donated to the analyzer group from both process and pilot plants. They had been deemed not needed for the production unit after sufficient process knowledge was gained. We could trial an analyzer application for only the installation and maintenance cost. Some production units would donate an analyzer if they found they could control the process with a more common and less costly measurement, such as pressure or temperature. Some production units refused to give them up even if the information was not being used. Another source of opportunities came from process engineers who saw a need or opportunity to better control their processes and came to us for assistance.
Greg: Where were analyzers essential for control?
Jim: Analyzers were needed on high-purity columns because temperature was not sensitive to the parts per million of impurities that needed to be controlled. In some cases, we were able to have several optional sample points installed that proved valuable because steady-state process simulations were not accurate enough to show the best tray for composition control. Another common application was fluidized bed reactors to optimize turnarounds for catalyst replacement. Before the analyzers were installed, the catalyst was changed out based on a time schedule, regardless of the degree of poisoning or decrease in activity of the catalyst. The result was the cost of the loss of yield and capacity when beds had unexpectedly deteriorated or the cost of catalyst and the loss of production when beds were still productive. In one case a $250K analyzer installation paid for itself in six months in cost savings.
Stan: What was your favorite analyzer?
Jim: If a sample was vaporizable, my first choice was usually the gas chromatograph (GC) because the fundamentals and maintenance were relatively simple. The optional detector and column configurations allowed us to frequently provide the process with a reliable, maintainable, specific and appropriate measurement. A technician could understand and maintain the GC. Often maintenance consisted of timing adjustments or simply replacing sample valves.
Greg: When did you have to pursue other analyzers?
Jim: When the sample wasn't vaporizable, had solids and bubbles, or required special sample preparation that would add excessive cost and time delay, we went for spectroscopic techniques. IR, UV and Fourier Transform Infrared (FTIR) had strong enough signals to use selective optical detectors. The weak signal from near-infrared (NIR) and Raman spectroscopy required chemometrics (data-driven multivariate statistical models), such as partial least squares or projection to latent structures (PLS) models. NIR and Raman analyzers have been developed for many applications and have the attractive feature of in situ probes advertised as eliminating the need for a sample system.