Interested in linking to "Long-Distance Calibration"?
You may use the Headline, Deck, Byline and URL of this article on your Web site. To link to this article, select and copy the HTML code below and paste it on your own Web site.
But Ceuta is no backwater. It is a very modern city with daily, high-speed ferry service to and from Cádiz in Spain. It is a free port, and oil, industry, retail and tourism drive its economy. Its population of some 78,000 gets its electricity from the Spanish electric power provider Endesa Generación.
Endesa generates 33% of Spain's electricity and has 25 plants spread over the Iberian Peninsula, plus five in the Balearic islands in the Mediterranean, nine in the Canary Islands, and two in Ceuta and Melilla, a similar city on the north coast of Africa near the Moroccan border. The power plant in Ceuta has an installed capacity of about 100 MW provided by nine diesel generators and a gas turbine.
See Also: Smart Calibration
Like most electricity providers today, Endesa's objective is to optimize its competitive position by producing electricity at a minimum cost, all while maximizing the availability of its equipment, ensuring the safety of its employees and respecting the natural environment. This is a tall order given that Endesa has electrical generating facilites spread out all over the Iberian peninsula, the southwestern Mediterannean and out in the Atlantic Ocean off the west coast of Africa.
Part of fulfilling this tall order means that keeping track of greenhouse gas emissions for regulatory and sustainability purposes is a big issue for Endesa. This effort is supported by Endress+Hauser's Promass flowmeters installed on its facilities' gensets. But installing flowmeters is only part of the solution. Those meters have to be calibrated, in Ceuta's case, every three years. And here's where things get a bit more complicated.Germán Canosa Murcia, electrical maintenance technician at Endesa in Ceuta, explains: "The calibration of these flowmeters is extremely important because first, the Spanish electricity system operator, Red Eléctrica of Spain SA, requires testing on certain flowmeters to ensure adequate performance, and these have to be calibrated every three years. We also have to properly calculate fuel consumption and CO2 emissions to comply with Spanish and European Union regulations. Finally, we have an internal directive of the company to keep track of these things." (Figure 1)
See Also: Save Money. Calibrate Less?
But regulatory and corporate demands aside, the calibration of the 11 flowmeters at the Ceuta facility present other challenges. All the flowmeters could not be removed at the same time because each genset has to be stopped when the flowmeter is removed, and Ceuta had only one backup flowmeter. Calibrating all of them at once would virtually shut down the facility. But doing them one at a time also presented problems. The process would drag out over nearly a year and incur extra costs.
Canosa says, "The original plan was to remove a flowmeter, replace it with a backup and send the flowmeter back to the manufacturer for calibration. But Ceuta has special tax and customs procedures, and the equipment must pass both procedures when being sent and when it is returned. This fact, coupled with the transporting and the calibration time, could take as long as a month for each one. With 11 flowmeters, the total calibration time could be up to 11 months."
At Ceuta, says Canosa: "First our maintenance people dismantled the flowmeters, and then cleaned and decontaminated them. Then the Endress+Hauser engineers connected the flowmeter to the testing computer, and checked and verified its calibration. If the calibration tests showed that the measurements were not as specified, the engineers recalibrated the system until the correct accuracy was achieved. The calibration takes between one and two hours. The entire process, including disassembly, assembly and calibration does not exceed more than three of four hours per flowmeter."
See Also: Market Intelligence Report: FlowmetersCanosa continues, "The advantages of on-site calibration are evident. In just four or five days, we had calibrated all the flowmeters. We also saved money by not having to pay for the transport and the special tax and customs procedures involved in shipping them back to the manufacturer."
Endesa has not limited its use of Endress+Hauser's services to Ceuta. The Coriolis flowmeters in service in its diesel plants in Melilla, just up the coast from Ceuta, on the Balearics in the Mediterranean, and in the Canary Islands, about 62 mi (100 km) off the coast of western Africa, have also benefited from the on-site calibration service.
As for Ceuta, Canosa says the facility is already planning to use the service again next year when the next three-year calibration is required. "This time we will expand the on-site services to include the calibration of the temperature sensors within our flowmeters. This will increase the service
cost, but it will guarantee us more accurate temperature measurements.