Fluke is voluntarily recalling its digital clamp meters - Fluke 373, 374, 375, and 376 - that were manufactured between Sept. 1, 2010, and Oct. 31, 2012.
The reason for the recall is that the printed circuit assembly in these units may not be properly fastened to the test lead input jack. This may result in inaccurate voltage readings, including a low- and no-voltage reading on a circuit energized with a hazardous voltage, presenting a shock, electrocution or thermal burn hazard.
This voluntary recall involves Fluke Digital Clamp Meters with the model and serial numbers listed below. Any serial number preceded or followed by the letter "R" is NOT affected by this recall notice.
|Model||Manufacture Dates||Starting Serial Number||Ending Serial Number|
|Fluke 376||Sept 2010 – Feb 2011||14270001||15909999|
|Fluke 375||Sept 2010 – Mar 2011||14270001||16079999|
|Fluke 374||Oct 2010 – Feb 2011||14270001||16379999|
|Fluke 373||Oct 2010 – Oct 2012||14270001||
Anyone who currently owns one of these clamp meters should stop using it immediately and send it back to Fluke for repair. When sending the device back, please send only the clamp meter. Any test leads/boxes/manuals that are sent with the unit will NOT be returned.
To contact Fluke for information on how to have your units repaired, either:
- Visit www.fluke.com/37x-recall to register your affected product online as part of the recall. Once your information is complete and you click "submit," a postage-paid mailing envelope will be sent to you within two business days. Use this envelope to return your clamp meter to Fluke for repair. Your clamp meter will be sent back to you when repairs are complete.
- Call 877-564-7180 for information and to request a postage-paid mailing envelope be sent to you.