How to Select a Pressure Sensor for a Specific Application
There are over two hundred pressure sensor suppliers around the world, offering products from a few dollars to thousands of dollars. A purchaser or engineer unfamiliar with pressure sensors can become overwhelmed with the price range, quality and options. The first step is to understand his/her application from the media being measured, to the desired electrical output for indication or control. The following is a guide through a variety of options to make a prudent decision.
Media is the most important item when selecting a pressure sensor for an application. Most sensor suppliers only sell sensors that are rated for benign environments such as clean, dry air. The next tier of suppliers will sell products that will handle mild environments through to difficult/corrosive environments. Clean water, steam, some forms of hydraulic oils and Freon can be considered mild environments. Difficult media tends to be corrosive liquids and gases such as hydrogen sulfide, hydrochloric acid, bleach, bromides, waste water and hydrogen. Wrongful selection of a pressure sensor can lead to catastrophic failure and serious injury. When unsure, ask the pressure sensor manufacturer to provide a chemical compatibility chart with their products. In fluidic systems, such as water and hydraulics, one must understand how the water hammer and pressure transients effect the pressure sensor.
Author: American Sensor Technologies | File Type: PDF
Find more white papers on Field Instrumentation : Pressure